Main Article Content
Background and Purposes: Sidestream capnograghy is a potentially lifeaving tool for monitoring the ventilation in patients receiving general anesthesia. The sample gas that withdraw continuously throughout respiratory cycle and sampled in an analyzer contained anesthetized gases. Without proper control, these released gases after analysis may stay diffused in the operating room, affecting the personnel's health condition. The purpose of this article was to invent the connector for using with anesthetic machine particularly thoes that having no part of sampling gas return port, in order to reduce the risk of pollution analyzed gases.
Method: To create a connector for delivering analyzed gases from the sidestream capnography back into the breathing circuit system of the anesthetic machine that does not have a sampling gas return port or for ventilating them for reuse.
Results: The anesthetic gas return connector is an invention used with anedthetic machine that does not have a sample gas return port for the improvement of anesthetic service quality. This invention leads to the significant decrease in the budget of operation.
Conclusion: The anesthetic gas return connector is a self-sufficient, inexpensive, practical and highly efficient invention that could be sterilized and reused. The anesthesiologist could safely use the sidestream capnography to monitor the patient in the operating room.
This work is licensed under a Creative Commons Attribution-NonCommercial-NoDerivatives 4.0 International License.
2. Moon RE, Camporesi EM. Respiratory monitoring. In: Miller RD, editor. Miller’s Anesthesia. 6th ed. Philadelphia: Churchill Livingstone 2005:1438-72.
3. Hagerty JJ, Kleinman ME, Zurakowski D, Lyons AC, Krauss B. Accuracy of a new low-flow sidestream capnography technology in newborns: a pilot study. J Perinatol 2002;22:219-25.
4. Kasuya Y, AkÇa O, Sessler DI, Ozaki M, Komatsu R. Accuracy of postoperative end tidal PCo2 measurements with mainstream and sidestream capnography in non-obese patients and in obese patients with and without obstructive sleep apnea. Anesthesiology 2009;111:609-15.
5. Greenvberg SB, Murphy GS, Vender JS. Standard Monitoring Techniques. In: Barash PG, Cullen BF, Stoelting RK, Cahalan MK, Stock MC, editors. Clinical anesthesia. 6th ed. Philadelphia: Lippincott Williams & Wilkins 2009:698-700.
6. Dorsch JA, Dorsch SE. Pulse Oximetry. In: Dorsch JA, Dorsch SE, editors. Understanding Anesthesia Equipment. 4th ed. Pennsylvania: Lippincott Williams & Wilkins 1994:811-47.
7. Ortega R, Connor C, Kim S, Djang R, Patel K. Monitoring ventilation with capnography. N Engl J Med 2012;367(Suppl):27.
8. Gilboy N. Hawkins MR. Noninvasive monitoring of end-tidal carbon dioxide in the emergency department. Adv Emerg Nurs J 2006;28:301-13.
9. Draeger Medical Inc. Brochure Vamos plus Anesthetic gas monitor [Internet]. [cited 2013 Jun 7]; [about 4 p.]. Available from: https://www.draeger.ae/media/10/00/92/10009228/vamos_br_9048961_en.pdf
10. Draeger Medical Inc. Fabius GS Operator’s Instruction Manual [Internet]; 2003. Chapter 11, Components; [cited 2013 Jun 7]; 119-46. Available from: https://etherweb.bwh.harvard.edu/education/PHILIP/Fabius_User_Manual.pdf
11. Jaffe MB. Mainstream or Sidestream Capnography? [Internet]. Respironics, Inc.; 2002 [cited 2013 Jun 6]. [about 14 p.]. Available from: https://www.oem.respironics.com/Downloads/Main vs. Side.pdf
12. Yagiela JA. Health hazards and nitrous oxide: a time for reappraisal. Anesth Prog 1991;38:1-11.
13. Maze M, Fujinaga M. Recent advances in understanding the actions and toxicity of nitrous oxide. Anaesthesia 2000;55:311-4.
14. Outstanding features of “Thai pipe” uPVC & Fittings [Internet]. Thai Pipe Industry Co., Ltd. [cited 2013 Oct 17]. Available from: https://www.thaipipe.com/product/view.php?cid=208.